地址：上海市闵行区兴梅路485号 xingmei rd 485,shanghai,china
显微镜物镜加热器 objective heater
temperature controlled heating band
easy installation around the objective
fits any objective on the market
compatible with the following okolab controllers:
compatible with inverted and upright microscopes
self calibrationroutines with bl controllers
voltage: 24 v
power: 13 w
when should i use an objective heater
you should use an objective heater, if you:
perform oil immersion imaging with a top stage incubator
use dipping lens with an upright top stage incubator
why should i use an objective heater?
because, if you don't, sample temperature will drop due to oil contact with the cold objective, as shown in the following graph.
step 1: objective not in contact with petri 2; objective heater off.
step 2: objective in oil contact with petri 2; objective heater off.
step 3: objective in oil contact with petri 2; objective heater on, offset value 5.3°c.
during the self calibration routine, the offset value of the objective heater is progressively increased, until t2 reaches the desired set point temperature of 37.0°c.
figure 3 reports the profile of t1, t2, and of objective heater temperature during the experimental campaign described above.